高低温试验箱ASTM G85-11标准来源：本站 发布时间：2018-11-06 19:05:00 阅读：1254
本页面标准信息均来源于网络收集，或由参与标准制定的供应商提供，只作为参考使用，为确保测试结果的准确性、节省您的时间，如需ASTM G85-11标准详细信息或ASTM G85-11具体测试方法，欢迎与我们技术人员联系或给我们留言，我们将为您相关的技术支持。
1.1 This practice covers and sets forth conditions for five modifications in salt spray (fog) testing for specification purposes. These are in chronological order of their development:
1.1.1 Annex A1, acetic acid-salt spray test, continuous.
1.1.2 Annex A2, cyclic acidified salt spray test.
1.1.3 Annex A3, seawater acidified test, cyclic (SWAAT).
1.1.4 Annex A4, SO2 salt spray test, cyclic.
1.1.5 Annex A5, dilute electrolyte cyclic fog dry test.
1.2 This practice does not prescribe the type of modification, test specimen or exposure periods to be used for a specific product, nor the interpretation to be given to the results.
1.3 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.3.1 Exception—Fahrenheit temperature values are given for information only throughout this practice.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to consult and establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents
B117 Practice for Operating Salt Spray (Fog) Apparatus
D609 Practice for Preparation of Cold-Rolled Steel Panels for Testing Paint, Varnish, Conversion Coatings, and Related Coating Products
D1141 Practice for the Preparation of Substitute Ocean Water
D1193 Specification for Reagent Water
D1654 Test Method for Evaluation of Painted or Coated Specimens Subjected to Corrosive Environments
E70 Test Method for pH of Aqueous Solutions With the Glass Electrode
Q-fog 盐雾腐蚀试验箱用于ASTM G85-11腐蚀测试的步骤
以上标准信息来自互联网搜索整理，如需了解完整的ASTM G85-11的信息或ASTM G85-11标准解读，欢迎详询。